Prisma E SEM
Talos F200i TEM
Explore MyScope Outreach
The 2016 Image Contest Grand Prize Winner
Learn how to engage in cryo-EM SPA
2017 Nobel Prize for Chemistry
Krios G3i Cryo-TEM
Webcast: How Image Accuracy Impacts Your Research
With the Peltier stage, humidity cycling experiments can be performed to characterize the sample's morphology and phase distribution at various relative humidity conditions.
Add to my Upgrades and Accessories List
Add Item and Request a Quote
Find another Accessory
When cooled between 0 and 10 °C, specimens of very diverse nature can be imaged and analyzed at relative humidity conditions up to 100% (typical chamber pressures required are in the range 3-10 mbar).
In-situ freeze-drying can be performed when the temperature range of the Peltier stage is set below 0 °C with a minimum of -20 °C. The temperature and the ramp-up profile are directly controlled through the Quanta user interface.
The Peltier Cooled Specimen Stage is used to maintain water on samples inside the ESEM specimen chamber. It uses a thermoelectric module to alter temperature, and this in conjunction with specimen chamber pressure creates condensation on the sample.
no items have been added